Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. In order to synchronize Channel A and B, you need to share a clock and a trigger. Write Voltage, Read Current. Also when i include negative voltages in my sweep say like if im sweeping from -2 to 2 then the current values in between go to e-8 range which cannot happen as my resitor value iss 2. What will this interface do? I didn’t check your program, but the waitcomplete command could help you. Break the problem into smaller steps and when you get stuck come back to this thread and let us know where you are stuck.
|Date Added:||5 June 2008|
|File Size:||64.23 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Start small and build on what you already know.
Anyway, I just want to simply to program the Keithley A to run a certain voltage through the Hi and Lo cables, and read the output current. Message 4 of 5.
keithley – Discussion Forums – National Instruments
My VISA’s version is 4. If you need assistance on changing the address, please contact service lakeshore.
Most Active Software Boards: Message 2 of 4. I just had to reconfigure the DUT.
Search Instrument Driver Network
I was trying to send commands that interfered with the measurement before it was done. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Message 2 of 5. Message 2 of Message 1 of I’d recommend using the RS port. I have Labview bit.
Message 7 of Please help as im stuck at the oint for many days. Message 3 of 4.
We really can’t help without some specifics. Without the device itself it is going to be quite hard to see this error and work out what might be causing it. With the second cable, you then connect the two instruments in series with the PC.
Increasing Keithley sample rat – Discussion Forums – National Instruments
Message 1 of You can also use a GPIB interface: Message 4 of Message 8 of I dont have much experience about this. Message 10 of I scanned through them, and most of them were too complicated labciew me to undestand.
It’s just a simple: Do you want an interface between the two devices? I was able to find a discussion forum regarding the error code, the example, and a Kethley 26xxa product that may be of use to you. I have keithey the IV curve im getting thus.
Auto-suggest helps you quickly narrow down your search ieithley by suggesting possible matches as you type. Message 4 of I used the Keithley example with the the driver to make the program. I would suggest you start small.